Upload the image somewhere else and provide a link. This example demonstrates how to synchronize pulsed sweeps between two SMU channels on a dual channel Series B SourceMeter instrument. Yes I use keithley labview driver. I couldn’t get your VI to download, but i did look at the specs for the device and it has better resolution the 1V or 1A:. Maybe do a reset before you try.. If I find something let you know.
|Date Added:||16 January 2007|
|File Size:||34.92 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
How can I add a knob on the system so that I can choose which buffer element to print according labvieq th value of the knob? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Message 3 of Message 5 of Couldn’t download your kithley. Message 6 of Most Active Hardware Boards: Email Required, but never shown.
Tektronix and Keithley
Hi Antme, This is tricky to troubleshoot since National Instruments does not support most 3rd party drivers. One function outputs pulsed voltage sweeps while the other performs pulsed current sweeps. You labvisw the knob control, wire the value to the number-to-string VI http: LabVIEW searches for that string and replaces it with the knob’s current value. The number of points in the sweep pulseWidth: This is not the limit used during the pulses.
I don’t have enough reputation points to add images.
Value of Numeric control in script, LabView – Stack Overflow
Have you used debugging to look at the string that is being sent to the instrument? Message 7 of Keithley cannot read decimal point of the computer. Yes, you will need to “write” the control to the instrument using SCPI commands labvied in documentation for the instrument. This example performs linear sweeps but can easily be modified to perform log or list sweeps.
The code can easily be modified to trigger off other sources as well. There is also useful link: There are two functions contained in this example script.
After calling this function you will need to press the TRIG button on the instrument’s front panel to trigger the sweep to start. Hi, it was almost 1 year ago.
Solved: source problem at mV and mA – Discussion Forums – National Instruments
Sure you can, please add the code you are using so we can help. Message 2 of Message 4 of Sign up using Email and Password.
The time from the start of one pulse to the start of the next pulse limitI: This example also programs both SMUs to source the same levels however, this can also easily be modified for each SMU ieithley output different levels.